NOTE: More detailed information about each project is on the Confluence pages.
SECTOR 5 LIST
|PROJECT||NEXT MILESTONE||DEADLINE||PREVIOUS MILESTONE|
|B4||Final implementation of b4
Update May 16: Beta version is running. All features now included. Processing results in GUI. Waiting for NE-CAT personnel to send RAPD program.
|April 28, 2017||All functionality implemented and tested Jan-Mar 2017. Available for users.|
|Vortex||Update March 20: Integrated into b3/b4, with excitation scans enabled.
Next step: label peaks in excitation scans.
|April 28, 2017||Integrated into b3/b4, with excitation scans enabled, and realtime control of deadtime. Available for users.|
|Beamline characterization||Develop tools for automatic COR detection
Next step: make available for users.
|May 17, 2017||Positions for Alignment jigs and VersaPins (position Z1) installed into robot dewars on 501 and 502. Robot mount/unmount tested and script for finding COR tested. Available for staff.|
|Investigate ways to make the robot faster
Update May 8: Testing of stationary collimator (in up position) completed.
|May 17, 2017||Investigation on 501 shows that clearance okay for collimator to remain in place during mount/unmount.|
Next step: testing of the automatic rastering (to determine crystal location in loop). Feature is available in b4, but not thoroughly tested yet.
|May 27, 2017||Automatic loop centering finished and available in b4. Next step: automatic rastering to located crystal in loop.|
COMPLETED SECTOR 5 ITEMS
|Gonio improvement||New snout||July 7, 2016||New snout ordered in May, arrived on June 13. Tested on June 13. Turned out to be slightly longer than old one, so adjusted gonio and robot accordingly.|
|Autoprocessing||mtz file creation||July 29, 2016||The autoprocessing script now includes output log and mtz files from Pointless and Aimless. and Also automatic processing independently with DIALS.|
|B4||Add ability for manual evaluation.||July 18, 2016||Will not be available in b3, but b4 can be run simultaneously with b3 to access this feature.|
|Reduction of ice in dewars||Copper-bottom lid installed on 501 to reduce friction between lid and dewar.||August 16, 2016||Installed and tested.|
|Pilatus – 502||Replaced 2 modules and associated electronic boards (new design from Dectris) and internal cable||August 18, 2016||This introduced a new error with the cancel run command. which Dectris fixed during January shutdown.|
|B3||Show SpotFinder results for every image||Sept 26, 2016||Also implemented in b4.|
|B3||Shutterless raster installed on 502.||Oct 7, 2016||Raster sped up on ADSC beamlines as well.|
|Centering stability||AC Unit for 503||Oct 26, 2016||Drain line feeds into ring area drain|
|Expanding capabilities||Kappa on 502||Nov 7, 2016||New interlocks installed to allow full rotation of phi when using kappa. Also modified DBS brcketry.|
|Beamline characterization/optimization||Automatic calibration of collimator size scripts implemented||Nov 30, 2016||Code is in python, and uses SciPy.|
|Expanding capabilities||Shutterless helical vector||Dec 14 2016||Note that shutterless raster and non-shutterless vector are already in operation|
|Pin base warmer||Installed on 5.0.1 and 5.0.2. Runs automatically between pins upon exchange. User has control over base heating as needed.||Mar 2017||Measured temperature at loop when pin base warmer is on. Temperature is stable for two minutes while pin base is warming.|
|Pilatus – 501||New Pilatus detector installed and tested on 5.0.1. Running simultaneously on same PPU as 5.0.2.||Mar 2017||Column of bad pixels in corner. Masked out for now. Dectris scheduled to fix during June 2-bunch.|
|Reduction of ice in dewars||New copper-base robot dewar lid on 5.0.2||Mar 2017||Installed and tested.|
|Dynamic beamstop||Installed on 5.0.1 (Mitegen version) and 5.0.2 (homemade version). Readout available in b4.||April 2017|
SECTOR 2 LIST
|PROJECT||NEXT MILESTONE||DATE||PREVIOUS MILESTONE|
|Monochromator||Onsite acceptance testing||June 2017||Vendor chosen; initial design complete. Seismic calculations done 6/27/16.
Vendor design review complete 7/28/16.
Final Design Review complete 9/19/16.
Seismic approval complete March 2017.
|Mirrors for primary beamline||Onsite acceptance testing||July 2017||Preliminary Design Review completed Oct 17. Final design review completed Dec 7, 2016. Seismic approval completed Mach 2017.|
|Undulator||Site visit (Chuck Swenson).||July 2017||Spec complete. Hitachi chosen as vendor. Preliminary design review completed. Projected arrival date: August 2017.|
|Hutch||Procurement using Sole Source||July 2017||CAD model complete. Scope and costing analysis complete 6/24/16. Review of design complete 9/4/2016. Bids received. Will order in July.|
JUNE POSSIBLE 2-BUNCH PROJECTS
|Fix kappa issue on 821 (see Jira BL821-309)||Jeff|
|Fix zoom1 motor on 821||Jeff|
|Use EPICS to build LabView-equivalent features for Gemini (eg, ability to scan slits or energy against diode reading)||John, Randall, Jeff, Kevin|
|Use EPICS on PowerPmac||Randall|
|Look into alternatives for SpotFinder||John|
|Fix 501 Pilatus pixels||Marc/Dectris|
|Independent control of shutter?|
REGULAR SCHEDULED MAINTENANCE
|ITEM||NEXT PLANNED||PREVIOUSLY DONE|
|Cryojet preventative maintenance – all beamlines||August, 2017||Jan 2017|
|501/503 mono flush||August, 2017||Feb 2017|
|All detector chillers – flush coolant||Jan-Feb, 2018||Jan 2016; Feb 2017|
|All detector chillers – vacuum filters||Jan, 2018||Jan, 2017|
|All detector PCs, ccd1 and ccd2 machines – vacuum filters||Jan 2018||Jan 2016; Jan 2017|
|502 cryocooler warmup||Jan 2018||April 2016; Feb 2017|
|821 cryocooler warmup||Jan 2018||April 2016; Feb 2017|
|Replace 501 robot SmartMotors for dewar||*|
|Replace 502 robot SmartMotors for dewar||*||Jan, 2017|
|Replace 503 robot SmartMotors for dewar||*||Jan, 2017|
|501 replace gripper||*||April, 2016|
|502 replace gripper||*||April, 2016|
|503 replace gripper||*||Aug, 2014|
|502 replace collimator||July 2019||July, 2016|
|502 replace goniometer snout||*||June, 2016|
|821/822 robot joints grease, J5 belt replace||Oct 2018||Oct, 2013|
* for these items, we have not yet determined the best schedule for maintenance/replacement.