The robot dewars have been taken off of 502 and 503 in order to replace the motors and perform maintenance on the stages.
Anthony installed the first pin base warmer on 5.0.2 today. We have to wait until the ALS has LN restored before we can test it. One thing to note: the people we spoke to at NE-CAT about their pin base warmer told us that it does not actually solve the icing issue completely. They still do see pins move on the goniometer after mounting. Hopefully the warmer will help a bit, though!
Marc collected a lysozyme dataset with the new Pilatus 6M detector. All modules are good.
A special welcome to our SULI student, Sayan Roychowhury, who is majoring in Electrical Engineering and Computer Science at the University of Illinois. He is here with us for the fall term, and writing code to process visual images of protein crystals and center them automatically in the x-ray beam. SULI is the Science Undergraduate Laboratory Internship program in which the host lab (in this case, LBNL) pays for a student to work for a few months to gain experience at a national lab and to encourage students to remain in STEM fields.
The compact dynamic beamstop (formerly known as the diode beamstop) has been announced as a finalist for this years R&D100 award! Simon Morton and Diane Bryant are invited to the Awards Banquet in Washington DC later this year.